From Open Energy Information
|Property Type|| Text
Pages using the property "UX-Recommendations"
Showing 11 pages using this property.
|MHK ISDB/Instruments/BlueView M900-2250 Dual Frequency 2D Imaging Sonar#UserExperience +||I've had very fast responses from their customer support team via phone and email. +|
|MHK ISDB/Instruments/Chelonia C-POD#UserExperience +||Caveat: haven't used since 2012. There were sensitivity differences between instruments mounted right next to each other. Calibration is recommended. +|
|MHK ISDB/Instruments/Futek TRS605 Rotary Torque Sensor#UserExperience +||The excitation voltage for the torque sensor and encoder portions of this sensor are different. Do not accidentally switch them. +|
|MHK ISDB/Instruments/Kongsberg M3 Multibeam Echosounder#UserExperience +||The software broadcasts the data stream, which can cause problems for other instruments operating on the same network. To allow the M3 data stream to be accessible by another computer, while still streaming data from other instruments (on a PC), I needed to allow a Windows Firewall exception for only the specific IP of the computer, as opposed to the entire M3 software package. +|
|MHK ISDB/Instruments/Loggerhead DSG-ST Ocean Acoustic Datalogger#UserExperience +||The continuous sampling rate was limited to 80 kHz, which was more than sufficient for my applications but may not be for others. Numerous changes have been made to the firmware etc since I last worked with the instrument. Some of the shortcomings may have since been addressed with these changes (syncing clock). For example, external LEDs on the hydrophone now indicate power to the instrument without opening the pressure housing. +|
|MHK ISDB/Instruments/Nortek Acoustic Doppler Velocimeter#UserExperience +||Always check the Nortek forums for answers to common questions. There is a large user base for these instruments and the forums are active. +|
|MHK ISDB/Instruments/Nortek Signature#UserExperience +||The Nortek help forum is a quick and easy way to get help with the instrument – Nortek typically will not respond to emails requesting tech support, but respond to forum posts very quickly! +, Caution about battery estimations (Although this might been fixed) +|
|MHK ISDB/Sensors/AmCells STA Series Load Cell#UserExperience +||If used for compression, calibrate with compressive loads. If used for tension, calibrate with tensile loads. +|
|MHK ISDB/Sensors/CR4100 Series AC Current Transducer#UserExperience +||Ensure the signal you're attempting to measure is within the frequency range (20 Hz – 5kHz) it can measure. If you're looking at PWM voltage (typically in the 10s of kHz) this sensor won’t give you a valid reading. +|
|MHK ISDB/Sensors/CR4500 Series AC Voltage Transducer#UserExperience +||Ensure the signal you're attempting to measure is within the frequency range (20 Hz – 5kHz) it can measure. If you're looking at PWM voltage (typically in the 10s of kHz) this sensor won’t give you a valid reading. +|
|MHK ISDB/Sensors/Futek TFF425 Reaction Torque Sensor#UserExperience +||Isolate this sensor from vibration for quality measurements +|